During optimization, the inverse planning system changes the weight of the beamlets by one intensity level (one level higher or lower), one beamlet at a time. After each change, a 3D dose matrix is calculated, and DVHs are generated for the target and critical structures. If the DVHs come closer to satisfying the planning goals, the change is accepted. Otherwise, it is rejected. To avoid local minima, the simulated annealing algorithm initially accepts some changes that make the dose distribution worse.
I Wish You Knew
Screening with Mammography: Looking at the Controversy
Blogs and Web Chats
OncoLink Blogs give our readers a chance to react to and comment on key cancer news topics and provides a forum for OncoLink Experts and readers to share opinions and learn from each other.
Facing a new cancer diagnosis or changing the course of your current treatment? Let our cancer nurses help you through!