- Since we have limited experience with IMRT and dynamic MLC, we continue to verify each plan using an ion chamber in a water-equivalent phantom.
- The planning system is used to cast the patients IM beams into the phantom and predict the ion chamber dose for a single fraction.
For our patient, measurement and calculation agreed to within 2%.
Time = 60 min. when combined with fluence map verification
I Wish You Knew
Screening with Mammography: Looking at the Controversy
Blogs and Web Chats
OncoLink Blogs give our readers a chance to react to and comment on key cancer news topics and provides a forum for OncoLink Experts and readers to share opinions and learn from each other.
Facing a new cancer diagnosis or changing the course of your current treatment? Let our cancer nurses help you through!